An intermediate circuit board serves as a central connection point for two stepper motors, their limit stops, and another connector for an 8-pin flat flex cable that connects to a third circuit board. This patented multidirectional approach to gathering light in a circle enables Spectrolino to read the target at any angle. This small board is nothing more than a way to mount and connect the SpectroLino. In my case, this occurred and appeared normal. This may also mean that we are not knowledgeable in this specific item.
|Date Added:||22 February 2017|
|File Size:||46.72 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
We will honor a return if the item s have been deemed to be defective during the warranty period. An intermediate circuit board serves as a central connection point for two stepper motors, their limit stops, and another connector for an 8-pin flat flex cable that connects to a third circuit board. Super high amount of views. With the small circuit board zpectroscan looking questionable, I thought it best to test the continuity spectroecan the mini-DIN connector to the 8-pin flat flex cable connector.
For monitor measurements, attach the Spectrolino directly to the screen using the suction cup supplied for this purpose.
Rotated X, B Targets. At this point I began thinking that the issue was with the SpectroScan table and not the SpectroLino, but I wanted confirmation that my SpectroLino was working. This may also mean that we are not knowledgeable in this specific item.
It also tilts the SpectroLino up and down as it maneuvers and takes measurements. Tried two different Keyspan serial-to-USB adapters.
Whatever the case, I was now acutely focused on this circuit as the root of my problem. In my case, this occurred and appeared normal.
X, B, I Targets.
What we do is apply a small amount of aerosol silicone to a clean rag, apply the cloth to the rails, and then wipe off the rails as much as possible.
These stainless steel rails which the Lino glides around on need to be very clean except for a very light coating of lubricant.
Retrieved from ” http: After re-installing the mini-DIN connector, I soldered a jumper wire on the back side of the board as shown below.
The only difference is that a Keyspan adapter was inserted between the last gretgamacbeth and the computer. Offers for similar machines. Indeed, I actually spoke with him many years ago regarding another issue with my SpectroScan. Search Offers Advanced search Dealers Auctions.
We except returns under the condition that there was no clear description informing the buyer of the third party description or condition. To get access to the rails you will need to remove the cover off the upper unit that moves forward gretagmacberh back across the top of the table. After my teacher and I worked on it for weeks, we eventually gave up.
Spectrolino Product Support
It worked, and I was able to successfully reset the SpectroLino, connect with MeasureTool, and take a gretgamacbeth reading. Returns We Except We except returns under the condition that there spectroscam no clear description informing the buyer of the third party description or condition. Tried resetting the SpectroLino as specrtoscan on page 17 of the User Guide. Otherwise, specgroscan serial communication was broken, then nothing would happen when the Start button is clicked.
When I choose these settings, the status is returned as OK. I also thought about the original error code: The front control panel circuit board mounts underneath the top surface of the SpectroScan table: We accept Paypal payments, your Paypal must be a confirmed address and we do not ship large packages to po boxes. I am looking forward to hearing from you.
Your message to the recipient. If booked within XX: This bodge wire directly connects pin 7 of the mini-DIN connector to the trace that continues to pin 7 of the flat flex cable connector. After testing the cables, the good news was that each circuit on both cables tested good for continuity.